1 | #pragma rtGlobals=3 // Use modern global access method and strict wave access. |
---|
2 | #pragma version=1.0 |
---|
3 | #pragma IgorVersion=6.1 |
---|
4 | |
---|
5 | |
---|
6 | |
---|
7 | // |
---|
8 | // functions to apply corrections to the detector panels |
---|
9 | // |
---|
10 | // these are meant to be called by the procedures that convert "raw" data to |
---|
11 | // "adjusted" or corrected data sets |
---|
12 | // |
---|
13 | // may be relocated in the future |
---|
14 | // |
---|
15 | |
---|
16 | |
---|
17 | |
---|
18 | // |
---|
19 | // detector dead time |
---|
20 | // |
---|
21 | // input is the data array (N tubes x M pixels) |
---|
22 | // input of N x 1 array of dead time values |
---|
23 | // |
---|
24 | // output is the corrected counts in data, overwriting the input data |
---|
25 | // |
---|
26 | // Note that the equation in Roe (eqn 2.15, p. 63) looks different, but it is really the |
---|
27 | // same old equation, just written in a more complex form. |
---|
28 | // |
---|
29 | // TODO |
---|
30 | // -- verify the direction of the tubes and indexing |
---|
31 | // x- decide on the appropriate functional form for the tubes |
---|
32 | // x- need count time as input |
---|
33 | // -- be sure I'm working in the right data folder |
---|
34 | // -- clean up when done |
---|
35 | // -- calculate + return the error contribution? |
---|
36 | // -- verify the error propagation |
---|
37 | Function V_DeadTimeCorrectionTubes(dataW,data_errW,dtW,ctTime) |
---|
38 | Wave dataW,data_errW,dtW |
---|
39 | Variable ctTime |
---|
40 | |
---|
41 | // do I count on the orientation as an input, or do I just figure it out on my own? |
---|
42 | String orientation |
---|
43 | Variable dimX,dimY |
---|
44 | dimX = DimSize(dataW,0) |
---|
45 | dimY = DimSize(dataw,1) |
---|
46 | if(dimX > dimY) |
---|
47 | orientation = "horizontal" |
---|
48 | else |
---|
49 | orientation = "vertical" |
---|
50 | endif |
---|
51 | |
---|
52 | // sum the counts in each tube and divide by time for total cr per tube |
---|
53 | Variable npt |
---|
54 | |
---|
55 | if(cmpstr(orientation,"vertical")==0) |
---|
56 | // this is data dimensioned as (Ntubes,Npix) |
---|
57 | |
---|
58 | MatrixOp/O sumTubes = sumRows(dataW) // n x 1 result |
---|
59 | sumTubes /= ctTime //now count rate per tube |
---|
60 | |
---|
61 | dataW[][] = dataW[p][q]/(1-sumTubes[p]*dtW[p]) //correct the data |
---|
62 | data_errW[][] = data_errW[p][q]/(1-sumTubes[p]*dtW[p]) // propagate the error wave |
---|
63 | |
---|
64 | elseif(cmpstr(orientation,"horizontal")==0) |
---|
65 | // this is data (horizontal) dimensioned as (Npix,Ntubes) |
---|
66 | |
---|
67 | MatrixOp/O sumTubes = sumCols(dataW) // 1 x m result |
---|
68 | sumTubes /= ctTime |
---|
69 | |
---|
70 | dataW[][] = dataW[p][q]/(1-sumTubes[q]*dtW[q]) |
---|
71 | data_errW[][] = data_errW[p][q]/(1-sumTubes[q]*dtW[q]) |
---|
72 | |
---|
73 | else |
---|
74 | DoAlert 0,"Orientation not correctly passed in DeadTimeCorrectionTubes(). No correction done." |
---|
75 | endif |
---|
76 | |
---|
77 | return(0) |
---|
78 | end |
---|
79 | |
---|
80 | // test function |
---|
81 | Function V_testDTCor() |
---|
82 | |
---|
83 | String detStr = "" |
---|
84 | String fname = "RAW" |
---|
85 | Variable ctTime |
---|
86 | |
---|
87 | detStr = "FR" |
---|
88 | Wave w = V_getDetectorDataW(fname,detStr) |
---|
89 | Wave w_err = V_getDetectorDataErrW(fname,detStr) |
---|
90 | Wave w_dt = V_getDetector_deadtime(fname,detStr) |
---|
91 | |
---|
92 | ctTime = V_getCount_time(fname) |
---|
93 | |
---|
94 | // ctTime = 10 |
---|
95 | V_DeadTimeCorrectionTubes(w,w_err,w_dt,ctTime) |
---|
96 | |
---|
97 | End |
---|
98 | |
---|
99 | |
---|
100 | // |
---|
101 | // Non-linear data correction |
---|
102 | // |
---|
103 | // DOES NOT modify the data, only calculates the spatial relationship |
---|
104 | // |
---|
105 | // input is the data array (N tubes x M pixels) |
---|
106 | // input of N x M array of quadratic coefficients |
---|
107 | // |
---|
108 | // output is wave of corrected real space distance corresponding to each pixel of the data |
---|
109 | // |
---|
110 | // |
---|
111 | // TODO |
---|
112 | // -- UNITS!!!! currently this is mm, which certainly doesn't match anything else!!! |
---|
113 | // -- verify the direction of the tubes and indexing |
---|
114 | // -- be sure I'm working in the right data folder |
---|
115 | // -- clean up when done |
---|
116 | // -- calculate + return the error contribution? |
---|
117 | // -- do I want this to return a wave? |
---|
118 | // -- do I need to write a separate function that returns the distance wave for later calculations? |
---|
119 | // -- do I want to make the distance array 3D to keep the x and y dims together? Calculate them all right now? |
---|
120 | // -- what else do I need to pass to the function? (fname=folder? detStr?) |
---|
121 | // -- need a separate block or function to handle "B" detector which will be ? different |
---|
122 | // |
---|
123 | // |
---|
124 | Function V_NonLinearCorrection(dataW,coefW,tube_width,detStr,destPath) |
---|
125 | Wave dataW,coefW |
---|
126 | Variable tube_width |
---|
127 | String detStr,destPath |
---|
128 | |
---|
129 | |
---|
130 | // do I count on the orientation as an input, or do I just figure it out on my own? |
---|
131 | String orientation |
---|
132 | Variable dimX,dimY |
---|
133 | dimX = DimSize(dataW,0) |
---|
134 | dimY = DimSize(dataW,1) |
---|
135 | if(dimX > dimY) |
---|
136 | orientation = "horizontal" |
---|
137 | else |
---|
138 | orientation = "vertical" |
---|
139 | endif |
---|
140 | |
---|
141 | // make a wave of the same dimensions, in the same data folder for the distance |
---|
142 | // ?? or a 3D wave? |
---|
143 | Make/O/D/N=(dimX,dimY) $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
144 | Make/O/D/N=(dimX,dimY) $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
145 | Wave data_realDistX = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
146 | Wave data_realDistY = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
147 | |
---|
148 | // then per tube, do the quadratic calculation to get the real space distance along the tube |
---|
149 | // the distance perpendicular to the tube is n*(8.4mm) per tube index |
---|
150 | |
---|
151 | if(cmpstr(orientation,"vertical")==0) |
---|
152 | // this is data dimensioned as (Ntubes,Npix) |
---|
153 | data_realDistX[][] = tube_width*p |
---|
154 | data_realDistY[][] = coefW[0][p] + coefW[1][p]*q + coefW[2][p]*q*q |
---|
155 | |
---|
156 | elseif(cmpstr(orientation,"horizontal")==0) |
---|
157 | // this is data (horizontal) dimensioned as (Npix,Ntubes) |
---|
158 | data_realDistX[][] = coefW[0][q] + coefW[1][q]*p + coefW[2][q]*p*p |
---|
159 | data_realDistY[][] = tube_width*q |
---|
160 | |
---|
161 | else |
---|
162 | DoAlert 0,"Orientation not correctly passed in NonLinearCorrection(). No correction done." |
---|
163 | endif |
---|
164 | |
---|
165 | return(0) |
---|
166 | end |
---|
167 | |
---|
168 | // TODO: |
---|
169 | // -- the cal_x and y coefficients are totally fake |
---|
170 | // -- the wave assignment may not be correct.. so beware |
---|
171 | // |
---|
172 | // |
---|
173 | Function V_NonLinearCorrection_B(folder,detStr,destPath) |
---|
174 | String folder,detStr,destPath |
---|
175 | |
---|
176 | if(cmpstr(detStr,"B") != 0) |
---|
177 | return(0) |
---|
178 | endif |
---|
179 | |
---|
180 | // do I count on the orientation as an input, or do I just figure it out on my own? |
---|
181 | Variable dimX,dimY |
---|
182 | |
---|
183 | Wave dataW = V_getDetectorDataW(folder,detStr) |
---|
184 | |
---|
185 | dimX = DimSize(dataW,0) |
---|
186 | dimY = DimSize(dataW,1) |
---|
187 | |
---|
188 | // make a wave of the same dimensions, in the same data folder for the distance |
---|
189 | // ?? or a 3D wave? |
---|
190 | Make/O/D/N=(dimX,dimY) $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
191 | Make/O/D/N=(dimX,dimY) $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
192 | Wave data_realDistX = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
193 | Wave data_realDistY = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
194 | |
---|
195 | |
---|
196 | Wave cal_x = V_getDet_cal_x(folder,detStr) |
---|
197 | Wave cal_y = V_getDet_cal_y(folder,detStr) |
---|
198 | |
---|
199 | data_realDistX[][] = cal_x[0]*p |
---|
200 | data_realDistY[][] = cal_y[0]*q |
---|
201 | |
---|
202 | return(0) |
---|
203 | end |
---|
204 | |
---|
205 | |
---|
206 | // |
---|
207 | // |
---|
208 | // TODO |
---|
209 | // -- VERIFY the calculations |
---|
210 | // -- verify where this needs to be done (if the beam center is changed) |
---|
211 | // -- then the q-calculation needs to be re-done |
---|
212 | // -- the position along the tube length is referenced to tube[0], for no particular reason |
---|
213 | // It may be better to take an average? but [0] is an ASSUMPTION |
---|
214 | // -- distance along tube is simple interpolation, or do I use the coefficients to |
---|
215 | // calculate the actual value |
---|
216 | // |
---|
217 | // -- distance in the lateral direction is based on tube width, which is a fixed parameter |
---|
218 | // |
---|
219 | // |
---|
220 | Function V_ConvertBeamCtr_to_mm(folder,detStr,destPath) |
---|
221 | String folder,detStr,destPath |
---|
222 | |
---|
223 | Wave data_realDistX = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
224 | Wave data_realDistY = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
225 | |
---|
226 | String orientation |
---|
227 | Variable dimX,dimY,xCtr,yCtr |
---|
228 | dimX = DimSize(data_realDistX,0) |
---|
229 | dimY = DimSize(data_realDistX,1) |
---|
230 | if(dimX > dimY) |
---|
231 | orientation = "horizontal" |
---|
232 | else |
---|
233 | orientation = "vertical" |
---|
234 | endif |
---|
235 | |
---|
236 | xCtr = V_getDet_beam_center_x(folder,detStr) |
---|
237 | yCtr = V_getDet_beam_center_y(folder,detStr) |
---|
238 | |
---|
239 | Make/O/D/N=1 $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_x_mm") |
---|
240 | Make/O/D/N=1 $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_y_mm") |
---|
241 | WAVE x_mm = $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_x_mm") |
---|
242 | WAVE y_mm = $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_y_mm") |
---|
243 | |
---|
244 | Variable tube_width = V_getDet_tubeWidth(folder,detStr) |
---|
245 | |
---|
246 | // |
---|
247 | if(cmpstr(orientation,"vertical")==0) |
---|
248 | // this is data dimensioned as (Ntubes,Npix) |
---|
249 | // data_realDistX[][] = tube_width*p |
---|
250 | // data_realDistY[][] = coefW[0][p] + coefW[1][p]*q + coefW[2][p]*q*q |
---|
251 | x_mm[0] = tube_width*xCtr |
---|
252 | y_mm[0] = data_realDistY[0][yCtr] |
---|
253 | else |
---|
254 | // this is data (horizontal) dimensioned as (Npix,Ntubes) |
---|
255 | // data_realDistX[][] = coefW[0][q] + coefW[1][q]*p + coefW[2][q]*p*p |
---|
256 | // data_realDistY[][] = tube_width*q |
---|
257 | x_mm[0] = data_realDistX[xCtr][0] |
---|
258 | y_mm[0] = tube_width*yCtr |
---|
259 | endif |
---|
260 | |
---|
261 | return(0) |
---|
262 | end |
---|
263 | |
---|
264 | |
---|
265 | // |
---|
266 | // |
---|
267 | // TODO |
---|
268 | // -- VERIFY the calculations |
---|
269 | // -- verify where this needs to be done (if the beam center is changed) |
---|
270 | // -- then the q-calculation needs to be re-done |
---|
271 | // -- the position along the tube length is referenced to tube[0], for no particular reason |
---|
272 | // It may be better to take an average? but [0] is an ASSUMPTION |
---|
273 | // -- distance along tube is simple interpolation |
---|
274 | // |
---|
275 | // -- distance in the lateral direction is based on tube width, which is well known |
---|
276 | // |
---|
277 | // |
---|
278 | Function V_ConvertBeamCtr_to_mmB(folder,detStr,destPath) |
---|
279 | String folder,detStr,destPath |
---|
280 | |
---|
281 | Wave data_realDistX = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
282 | Wave data_realDistY = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
283 | |
---|
284 | Variable xCtr,yCtr |
---|
285 | xCtr = V_getDet_beam_center_x(folder,detStr) |
---|
286 | yCtr = V_getDet_beam_center_y(folder,detStr) |
---|
287 | |
---|
288 | Make/O/D/N=1 $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_x_mm") |
---|
289 | Make/O/D/N=1 $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_y_mm") |
---|
290 | WAVE x_mm = $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_x_mm") |
---|
291 | WAVE y_mm = $(destPath + ":entry:instrument:detector_"+detStr+":beam_center_y_mm") |
---|
292 | |
---|
293 | x_mm[0] = data_realDistX[xCtr][0] |
---|
294 | y_mm[0] = data_realDistY[0][yCtr] |
---|
295 | |
---|
296 | return(0) |
---|
297 | end |
---|
298 | |
---|
299 | |
---|
300 | |
---|
301 | |
---|
302 | |
---|
303 | |
---|
304 | ///// |
---|
305 | // |
---|
306 | // non-linear corrections to the tube pixels |
---|
307 | // - returns the distance in mm (although this may change) |
---|
308 | // |
---|
309 | // c0,c1,c2,pix |
---|
310 | // c0-c2 are the fit coefficients |
---|
311 | // pix is the test pixel |
---|
312 | // |
---|
313 | // returns the distance in mm (relative to ctr pixel) |
---|
314 | // ctr is the center pixel, as defined when fitting to quadratic was done |
---|
315 | // |
---|
316 | Function V_TubePixel_to_mm(c0,c1,c2,pix) |
---|
317 | Variable c0,c1,c2,pix |
---|
318 | |
---|
319 | Variable dist |
---|
320 | dist = c0 + c1*pix + c2*pix*pix |
---|
321 | |
---|
322 | return(dist) |
---|
323 | End |
---|
324 | // |
---|
325 | //// |
---|
326 | |
---|
327 | // TODO |
---|
328 | // get rid of this in the real data |
---|
329 | // |
---|
330 | // TESTING ONLY |
---|
331 | Proc V_MakeFakeCalibrationWaves() |
---|
332 | // make these in the RAW data folder, before converting to a work folder |
---|
333 | // - then they will be "found" by get() |
---|
334 | // -- only for the tube, not the Back det |
---|
335 | |
---|
336 | // DoAlert 0, "re-do this and do a better job of filling the fake calibration data" |
---|
337 | |
---|
338 | DoAlert 0, "Calibration waves are read in from the data file" |
---|
339 | |
---|
340 | // V_fMakeFakeCalibrationWaves() |
---|
341 | End |
---|
342 | |
---|
343 | |
---|
344 | |
---|
345 | // TODO |
---|
346 | // get rid of this in the real data |
---|
347 | // |
---|
348 | // TESTING ONLY |
---|
349 | // |
---|
350 | // orientation does not matter, there are 48 tubes in each bank |
---|
351 | // so dimension (3,48) for everything. |
---|
352 | // |
---|
353 | // -- but the orientation does indicate TB vs LR, which has implications for |
---|
354 | // the (fictional) dimension of the pixel along the tube axis, at least as far |
---|
355 | // as for making the fake coefficients. |
---|
356 | // |
---|
357 | Function V_fMakeFakeCalibrationWaves() |
---|
358 | |
---|
359 | Variable ii,pixSize |
---|
360 | String detStr,fname="RAW",orientation |
---|
361 | |
---|
362 | for(ii=0;ii<ItemsInList(ksDetectorListNoB);ii+=1) |
---|
363 | detStr = StringFromList(ii, ksDetectorListNoB, ";") |
---|
364 | // Wave w = V_getDetectorDataW(fname,detStr) |
---|
365 | Make/O/D/N=(3,48) $("root:Packages:NIST:VSANS:RAW:entry:instrument:detector_"+detStr+":spatial_calibration") |
---|
366 | Wave calib = $("root:Packages:NIST:VSANS:RAW:entry:instrument:detector_"+detStr+":spatial_calibration") |
---|
367 | // !!!! this overwrites what is there |
---|
368 | |
---|
369 | orientation = V_getDet_tubeOrientation(fname,detStr) |
---|
370 | if(cmpstr(orientation,"vertical")==0) |
---|
371 | // this is vertical tube data dimensioned as (Ntubes,Npix) |
---|
372 | pixSize = 8.4 //V_getDet_y_pixel_size(fname,detStr) |
---|
373 | |
---|
374 | elseif(cmpstr(orientation,"horizontal")==0) |
---|
375 | // this is data (horizontal) dimensioned as (Npix,Ntubes) |
---|
376 | pixSize = 4 //V_getDet_x_pixel_size(fname,detStr) |
---|
377 | |
---|
378 | else |
---|
379 | DoAlert 0,"Orientation not correctly passed in NonLinearCorrection(). No correction done." |
---|
380 | endif |
---|
381 | |
---|
382 | calib[0][] = -(128/2)*pixSize //approx (n/2)*pixSixe |
---|
383 | calib[1][] = pixSize |
---|
384 | calib[2][] = 2e-4 |
---|
385 | |
---|
386 | endfor |
---|
387 | |
---|
388 | // now fake calibration for "B" |
---|
389 | Wave cal_x = V_getDet_cal_x("RAW","B") |
---|
390 | Wave cal_y = V_getDet_cal_y("RAW","B") |
---|
391 | |
---|
392 | cal_x = 1 // mm, ignore the other 2 values |
---|
393 | cal_y = 1 // mm |
---|
394 | return(0) |
---|
395 | End |
---|
396 | |
---|
397 | // |
---|
398 | // TODO: |
---|
399 | // -- MUST VERIFY the definition of SDD and how (if) setback is written to the data files |
---|
400 | // -- currently I'm assuming that the SDD is the "nominal" value which is correct for the |
---|
401 | // L/R panels, but is not correct for the T/B panels (must add in the setback) |
---|
402 | // |
---|
403 | // |
---|
404 | // |
---|
405 | // data_realDistX, Y must be previously generated from running NonLinearCorrection() |
---|
406 | // |
---|
407 | // call with: |
---|
408 | // fname as the WORK folder, "RAW" |
---|
409 | // detStr = detector String, "FL" |
---|
410 | // destPath = path to destination WORK folder ("root:Packages:NIST:VSANS:"+folder) |
---|
411 | // |
---|
412 | Function V_Detector_CalcQVals(fname,detStr,destPath) |
---|
413 | String fname,detStr,destPath |
---|
414 | |
---|
415 | String orientation |
---|
416 | Variable xCtr,yCtr,lambda,sdd |
---|
417 | |
---|
418 | // get all of the geometry information |
---|
419 | orientation = V_getDet_tubeOrientation(fname,detStr) |
---|
420 | // sdd = V_getDet_distance(fname,detStr) //[cm] |
---|
421 | // sdd += V_getDet_TBSetback(fname,detStr)/10 // written in [mm], convert to [cm], returns 0 for L/R/B panels |
---|
422 | |
---|
423 | sdd = V_getDet_ActualDistance(fname,detStr) //sdd derived, including setback [cm] |
---|
424 | sdd/=100 // sdd reported in cm, pass in m |
---|
425 | // this is the ctr in pixels |
---|
426 | // xCtr = V_getDet_beam_center_x(fname,detStr) |
---|
427 | // yCtr = V_getDet_beam_center_y(fname,detStr) |
---|
428 | // this is ctr in mm |
---|
429 | xCtr = V_getDet_beam_center_x_mm(fname,detStr) |
---|
430 | yCtr = V_getDet_beam_center_y_mm(fname,detStr) |
---|
431 | lambda = V_getWavelength(fname) |
---|
432 | Wave data_realDistX = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistX") |
---|
433 | Wave data_realDistY = $(destPath + ":entry:instrument:detector_"+detStr+":data_realDistY") |
---|
434 | |
---|
435 | // make the new waves |
---|
436 | Duplicate/O data_realDistX $(destPath + ":entry:instrument:detector_"+detStr+":qTot_"+detStr) |
---|
437 | Duplicate/O data_realDistX $(destPath + ":entry:instrument:detector_"+detStr+":qx_"+detStr) |
---|
438 | Duplicate/O data_realDistX $(destPath + ":entry:instrument:detector_"+detStr+":qy_"+detStr) |
---|
439 | Duplicate/O data_realDistX $(destPath + ":entry:instrument:detector_"+detStr+":qz_"+detStr) |
---|
440 | Wave qTot = $(destPath + ":entry:instrument:detector_"+detStr+":qTot_"+detStr) |
---|
441 | Wave qx = $(destPath + ":entry:instrument:detector_"+detStr+":qx_"+detStr) |
---|
442 | Wave qy = $(destPath + ":entry:instrument:detector_"+detStr+":qy_"+detStr) |
---|
443 | Wave qz = $(destPath + ":entry:instrument:detector_"+detStr+":qz_"+detStr) |
---|
444 | |
---|
445 | // calculate all of the q-values |
---|
446 | qTot = V_CalcQval(p,q,xCtr,yCtr,sdd,lambda,data_realDistX,data_realDistY) |
---|
447 | qx = V_CalcQX(p,q,xCtr,yCtr,sdd,lambda,data_realDistX,data_realDistY) |
---|
448 | qy = V_CalcQY(p,q,xCtr,yCtr,sdd,lambda,data_realDistX,data_realDistY) |
---|
449 | qz = V_CalcQZ(p,q,xCtr,yCtr,sdd,lambda,data_realDistX,data_realDistY) |
---|
450 | |
---|
451 | |
---|
452 | return(0) |
---|
453 | End |
---|
454 | |
---|
455 | |
---|
456 | //function to calculate the overall q-value, given all of the necesary trig inputs |
---|
457 | // |
---|
458 | // TODO: |
---|
459 | // -- verify the calculation (accuracy - in all input conditions) |
---|
460 | // -- verify the units of everything here, it's currently all jumbled and wrong... and repeated |
---|
461 | // -- the input data_realDistX and Y are essentially lookup tables of the real space distance corresponding |
---|
462 | // to each pixel |
---|
463 | // |
---|
464 | //sdd is in meters |
---|
465 | //wavelength is in Angstroms |
---|
466 | // |
---|
467 | //returned magnitude of Q is in 1/Angstroms |
---|
468 | // |
---|
469 | Function V_CalcQval(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
470 | Variable xaxval,yaxval,xctr,yctr,sdd,lam |
---|
471 | Wave distX,distY |
---|
472 | |
---|
473 | Variable dx,dy,qval,two_theta,dist |
---|
474 | |
---|
475 | sdd *=100 //convert to cm |
---|
476 | dx = (distX[xaxval][yaxval] - xctr) //delta x in mm |
---|
477 | dy = (distY[xaxval][yaxval] - yctr) //delta y in mm |
---|
478 | dist = sqrt(dx^2 + dy^2) |
---|
479 | |
---|
480 | dist /= 10 // convert mm to cm |
---|
481 | |
---|
482 | two_theta = atan(dist/sdd) |
---|
483 | |
---|
484 | qval = 4*Pi/lam*sin(two_theta/2) |
---|
485 | |
---|
486 | return qval |
---|
487 | End |
---|
488 | |
---|
489 | //calculates just the q-value in the x-direction on the detector |
---|
490 | // TODO: |
---|
491 | // -- verify the calculation (accuracy - in all input conditions) |
---|
492 | // -- verify the units of everything here, it's currently all jumbled and wrong... and repeated |
---|
493 | // -- the input data_realDistX and Y are essentially lookup tables of the real space distance corresponding |
---|
494 | // to each pixel |
---|
495 | // |
---|
496 | // |
---|
497 | // this properly accounts for qz |
---|
498 | // |
---|
499 | Function V_CalcQX(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
500 | Variable xaxval,yaxval,xctr,yctr,sdd,lam |
---|
501 | Wave distX,distY |
---|
502 | |
---|
503 | Variable qx,qval,phi,dx,dy,dist,two_theta |
---|
504 | |
---|
505 | qval = V_CalcQval(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
506 | |
---|
507 | sdd *=100 //convert to cm |
---|
508 | dx = (distX[xaxval][yaxval] - xctr) //delta x in mm |
---|
509 | dy = (distY[xaxval][yaxval] - yctr) //delta y in mm |
---|
510 | phi = V_FindPhi(dx,dy) |
---|
511 | |
---|
512 | //get scattering angle to project onto flat detector => Qr = qval*cos(theta) |
---|
513 | dist = sqrt(dx^2 + dy^2) |
---|
514 | dist /= 10 // convert mm to cm |
---|
515 | |
---|
516 | two_theta = atan(dist/sdd) |
---|
517 | |
---|
518 | qx = qval*cos(two_theta/2)*cos(phi) |
---|
519 | |
---|
520 | return qx |
---|
521 | End |
---|
522 | |
---|
523 | //calculates just the q-value in the y-direction on the detector |
---|
524 | // TODO: |
---|
525 | // -- verify the calculation (accuracy - in all input conditions) |
---|
526 | // -- verify the units of everything here, it's currently all jumbled and wrong... and repeated |
---|
527 | // -- the input data_realDistX and Y are essentially lookup tables of the real space distance corresponding |
---|
528 | // to each pixel |
---|
529 | // |
---|
530 | // |
---|
531 | // this properly accounts for qz |
---|
532 | // |
---|
533 | Function V_CalcQY(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
534 | Variable xaxval,yaxval,xctr,yctr,sdd,lam |
---|
535 | Wave distX,distY |
---|
536 | |
---|
537 | Variable qy,qval,phi,dx,dy,dist,two_theta |
---|
538 | |
---|
539 | qval = V_CalcQval(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
540 | |
---|
541 | sdd *=100 //convert to cm |
---|
542 | dx = (distX[xaxval][yaxval] - xctr) //delta x in mm |
---|
543 | dy = (distY[xaxval][yaxval] - yctr) //delta y in mm |
---|
544 | phi = V_FindPhi(dx,dy) |
---|
545 | |
---|
546 | //get scattering angle to project onto flat detector => Qr = qval*cos(theta) |
---|
547 | dist = sqrt(dx^2 + dy^2) |
---|
548 | dist /= 10 // convert mm to cm |
---|
549 | |
---|
550 | two_theta = atan(dist/sdd) |
---|
551 | |
---|
552 | qy = qval*cos(two_theta/2)*sin(phi) |
---|
553 | |
---|
554 | return qy |
---|
555 | End |
---|
556 | |
---|
557 | //calculates just the q-value in the z-direction on the detector |
---|
558 | // TODO: |
---|
559 | // -- verify the calculation (accuracy - in all input conditions) |
---|
560 | // -- verify the units of everything here, it's currently all jumbled and wrong... and repeated |
---|
561 | // -- the input data_realDistX and Y are essentially lookup tables of the real space distance corresponding |
---|
562 | // to each pixel |
---|
563 | // |
---|
564 | // not actually used for any calculations, but here for completeness if anyone asks, or for 2D data export |
---|
565 | // |
---|
566 | // this properly accounts for qz |
---|
567 | // |
---|
568 | Function V_CalcQZ(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
569 | Variable xaxval,yaxval,xctr,yctr,sdd,lam |
---|
570 | Wave distX,distY |
---|
571 | |
---|
572 | Variable qz,qval,phi,dx,dy,dist,two_theta |
---|
573 | |
---|
574 | qval = V_CalcQval(xaxval,yaxval,xctr,yctr,sdd,lam,distX,distY) |
---|
575 | |
---|
576 | sdd *=100 //convert to cm |
---|
577 | dx = (distX[xaxval][yaxval] - xctr) //delta x in mm |
---|
578 | dy = (distY[xaxval][yaxval] - yctr) //delta y in mm |
---|
579 | |
---|
580 | //get scattering angle to project onto flat detector => Qr = qval*cos(theta) |
---|
581 | dist = sqrt(dx^2 + dy^2) |
---|
582 | dist /= 10 // convert mm to cm |
---|
583 | |
---|
584 | two_theta = atan(dist/sdd) |
---|
585 | |
---|
586 | qz = qval*sin(two_theta/2) |
---|
587 | |
---|
588 | return qz |
---|
589 | End |
---|
590 | |
---|
591 | |
---|
592 | // |
---|
593 | // TODO -- VERIFY calculations |
---|
594 | // -- This is the actual solid angle per pixel, not a ratio vs. some "unit SA" |
---|
595 | // Do I just correct for the different area vs. the "nominal" central area? |
---|
596 | // -- decide how to implement - either directly change the data values (as was done in the past) |
---|
597 | // or use this as a weighting for when the data is binned to I(q). In the second method, 2D data |
---|
598 | // would need this to be applied before exporting |
---|
599 | // -- do I keep a wave note indicating that this correction has been applied to the data |
---|
600 | // so that it can be "un-applied"? |
---|
601 | // -- do I calculate theta from geometry directly, or get it from Q (Assuming it's present?) |
---|
602 | // (probably just from geometry, since I need SDD and dx and dy values...) |
---|
603 | // |
---|
604 | // |
---|
605 | Function V_SolidAngleCorrection(w,w_err,fname,detStr,destPath) |
---|
606 | Wave w,w_err |
---|
607 | String fname,detStr,destPath |
---|
608 | |
---|
609 | Variable sdd,xCtr,yCtr,lambda |
---|
610 | |
---|
611 | // get all of the geometry information |
---|
612 | // orientation = V_getDet_tubeOrientation(fname,detStr) |
---|
613 | sdd = V_getDet_ActualDistance(fname,detStr) |
---|
614 | sdd/=100 // sdd in cm, pass in m |
---|
615 | |
---|
616 | // this is ctr in mm |
---|
617 | xCtr = V_getDet_beam_center_x_mm(fname,detStr) |
---|
618 | yCtr = V_getDet_beam_center_y_mm(fname,detStr) |
---|
619 | lambda = V_getWavelength(fname) |
---|
620 | |
---|
621 | SetDataFolder $(destPath + ":entry:instrument:detector_"+detStr) |
---|
622 | |
---|
623 | Wave data_realDistX = data_realDistX |
---|
624 | Wave data_realDistY = data_realDistY |
---|
625 | |
---|
626 | Duplicate/O w solid_angle,tmp_theta,tmp_dist //in the current df |
---|
627 | |
---|
628 | //// calculate the scattering angle |
---|
629 | // dx = (distX - xctr) //delta x in mm |
---|
630 | // dy = (distY - yctr) //delta y in mm |
---|
631 | tmp_dist = sqrt((data_realDistX - xctr)^2 + (data_realDistY - yctr)^2) |
---|
632 | |
---|
633 | tmp_dist /= 10 // convert mm to cm |
---|
634 | sdd *=100 //convert to cm |
---|
635 | |
---|
636 | tmp_theta = atan(tmp_dist/sdd) //this is two_theta, the scattering angle |
---|
637 | |
---|
638 | Variable ii,jj,numx,numy,dx,dy |
---|
639 | numx = DimSize(tmp_theta,0) |
---|
640 | numy = DimSize(tmp_theta,1) |
---|
641 | |
---|
642 | for(ii=0 ;ii<numx;ii+=1) |
---|
643 | for(jj=0;jj<numy;jj+=1) |
---|
644 | |
---|
645 | if(ii==0) //do a forward difference if ii==0 |
---|
646 | dx = (data_realDistX[ii+1][jj] - data_realDistX[ii][jj]) //delta x for the pixel |
---|
647 | else |
---|
648 | dx = (data_realDistX[ii][jj] - data_realDistX[ii-1][jj]) //delta x for the pixel |
---|
649 | endif |
---|
650 | |
---|
651 | |
---|
652 | if(jj==0) |
---|
653 | dy = (data_realDistY[ii][jj+1] - data_realDistY[ii][jj]) //delta y for the pixel |
---|
654 | else |
---|
655 | dy = (data_realDistY[ii][jj] - data_realDistY[ii][jj-1]) //delta y for the pixel |
---|
656 | endif |
---|
657 | |
---|
658 | dx /= 10 |
---|
659 | dy /= 10 // convert mm to cm (since sdd is in cm) |
---|
660 | solid_angle[ii][jj] = dx*dy //this is in cm^2 |
---|
661 | endfor |
---|
662 | endfor |
---|
663 | |
---|
664 | // to cover up any issues w/negative dx or dy |
---|
665 | solid_angle = abs(solid_angle) |
---|
666 | |
---|
667 | // solid_angle correction |
---|
668 | // == dx*dy*cos^3/sdd^2 |
---|
669 | solid_angle *= (cos(tmp_theta))^3 |
---|
670 | solid_angle /= sdd^2 |
---|
671 | |
---|
672 | // Here it is! Apply the correction to the intensity (I divide -- to get the counts per solid angle!!) |
---|
673 | w /= solid_angle |
---|
674 | |
---|
675 | |
---|
676 | // TODO: |
---|
677 | // correctly apply the correction to the error wave (assume a perfect value?) |
---|
678 | // w_err /= solid_angle //is this correct?? |
---|
679 | |
---|
680 | // TODO -- clean up after I'm satisfied computations are correct |
---|
681 | // KillWaves/Z tmp_theta,tmp_dist |
---|
682 | |
---|
683 | return(0) |
---|
684 | end |
---|
685 | |
---|
686 | |
---|
687 | //////////// |
---|
688 | // TODO: all of below is untested code |
---|
689 | // copied from SANS |
---|
690 | // |
---|
691 | // |
---|
692 | // TODO : |
---|
693 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
694 | // |
---|
695 | //performs solid angle and non-linear detector corrections to raw data as it is "added" to a work folder |
---|
696 | //function is called by Raw_to_work() and Add_raw_to_work() functions |
---|
697 | //works on the actual data array, assumes that is is already on LINEAR scale |
---|
698 | // |
---|
699 | Function V_DetCorr(data,data_err,realsread,doEfficiency,doTrans) |
---|
700 | Wave data,data_err,realsread |
---|
701 | Variable doEfficiency,doTrans |
---|
702 | |
---|
703 | DoAlert 0,"This has not yet been updated for VSANS" |
---|
704 | |
---|
705 | Variable xcenter,ycenter,x0,y0,sx,sx3,sy,sy3,xx0,yy0 |
---|
706 | Variable ii,jj,dtdist,dtdis2 |
---|
707 | Variable xi,xd,yd,rad,ratio,domega,xy |
---|
708 | Variable lambda,trans,trans_err,lat_err,tmp_err,lat_corr |
---|
709 | |
---|
710 | // Print "...doing jacobian and non-linear corrections" |
---|
711 | |
---|
712 | NVAR pixelsX = root:myGlobals:gNPixelsX |
---|
713 | NVAR pixelsY = root:myGlobals:gNPixelsY |
---|
714 | |
---|
715 | //set up values to send to auxiliary trig functions |
---|
716 | xcenter = pixelsX/2 + 0.5 // == 64.5 for 128x128 Ordela |
---|
717 | ycenter = pixelsY/2 + 0.5 // == 64.5 for 128x128 Ordela |
---|
718 | |
---|
719 | x0 = realsread[16] |
---|
720 | y0 = realsread[17] |
---|
721 | sx = realsread[10] |
---|
722 | sx3 = realsread[11] |
---|
723 | sy = realsread[13] |
---|
724 | sy3 = realsread[14] |
---|
725 | |
---|
726 | dtdist = 1000*realsread[18] //sdd in mm |
---|
727 | dtdis2 = dtdist^2 |
---|
728 | |
---|
729 | lambda = realsRead[26] |
---|
730 | trans = RealsRead[4] |
---|
731 | trans_err = RealsRead[41] //new, March 2011 |
---|
732 | |
---|
733 | |
---|
734 | //waves to contain repeated function calls |
---|
735 | Make/O/N=(pixelsX) fyy,xx,yy //Assumes square detector !!! |
---|
736 | ii=0 |
---|
737 | do |
---|
738 | xi = ii |
---|
739 | // fyy[ii] = dc_fy(ii+1,sy,sy3,ycenter) |
---|
740 | // xx[ii] = dc_fxn(ii+1,sx,sx3,xcenter) |
---|
741 | // yy[ii] = dc_fym(ii+1,sy,sy3,ycenter) |
---|
742 | ii+=1 |
---|
743 | while(ii<pixelsX) |
---|
744 | |
---|
745 | Make/O/N=(pixelsX,pixelsY) SolidAngle // testing only |
---|
746 | |
---|
747 | ii=0 |
---|
748 | do |
---|
749 | xi = ii |
---|
750 | // xd = dc_fx(ii+1,sx,sx3,xcenter)-xx0 |
---|
751 | jj=0 |
---|
752 | do |
---|
753 | yd = fyy[jj]-yy0 |
---|
754 | //rad is the distance of pixel ij from the sample |
---|
755 | //domega is the ratio of the solid angle of pixel ij versus center pixel |
---|
756 | // product xy = 1 for a detector with a linear spatial response (modern Ordela) |
---|
757 | // solid angle calculated, dW^3 >=1, so multiply data to raise measured values to correct values. |
---|
758 | rad = sqrt(dtdis2 + xd^2 + yd^2) |
---|
759 | domega = rad/dtdist |
---|
760 | ratio = domega^3 |
---|
761 | xy = xx[ii]*yy[jj] |
---|
762 | |
---|
763 | data[ii][jj] *= xy*ratio |
---|
764 | |
---|
765 | solidAngle[ii][jj] = xy*ratio //testing only |
---|
766 | data_err[ii][jj] *= xy*ratio //error propagation assumes that SA and Jacobian are exact, so simply scale error |
---|
767 | |
---|
768 | |
---|
769 | // correction factor for detector efficiency JBG memo det_eff_cor2.doc 3/20/07 |
---|
770 | // correction inserted 11/2007 SRK |
---|
771 | // large angle detector efficiency is >= 1 and will "bump up" the measured value at the highest angles |
---|
772 | // so divide here to get the correct answer (5/22/08 SRK) |
---|
773 | if(doEfficiency) |
---|
774 | // data[ii][jj] /= DetEffCorr(lambda,dtdist,xd,yd) |
---|
775 | // data_err[ii][jj] /= DetEffCorr(lambda,dtdist,xd,yd) |
---|
776 | // solidAngle[ii][jj] /= DetEffCorr(lambda,dtdist,xd,yd) //testing only |
---|
777 | endif |
---|
778 | |
---|
779 | // large angle transmission calculation is <= 1 and will "bump down" the measured value at the highest angles |
---|
780 | // so divide here to get the correct answer |
---|
781 | if(doTrans) |
---|
782 | |
---|
783 | if(trans<0.1 && ii==0 && jj==0) |
---|
784 | Print "***transmission is less than 0.1*** and is a significant correction" |
---|
785 | endif |
---|
786 | |
---|
787 | if(trans==0) |
---|
788 | if(ii==0 && jj==0) |
---|
789 | Print "***transmission is ZERO*** and has been reset to 1.0 for the averaging calculation" |
---|
790 | endif |
---|
791 | trans = 1 |
---|
792 | endif |
---|
793 | |
---|
794 | // pass in the transmission error, and the error in the correction is returned as the last parameter |
---|
795 | lat_corr = V_LargeAngleTransmissionCorr(trans,dtdist,xd,yd,trans_err,lat_err) //moved from 1D avg SRK 11/2007 |
---|
796 | data[ii][jj] /= lat_corr //divide by the correction factor |
---|
797 | // |
---|
798 | // |
---|
799 | // |
---|
800 | // relative errors add in quadrature |
---|
801 | tmp_err = (data_err[ii][jj]/lat_corr)^2 + (lat_err/lat_corr)^2*data[ii][jj]*data[ii][jj]/lat_corr^2 |
---|
802 | tmp_err = sqrt(tmp_err) |
---|
803 | |
---|
804 | data_err[ii][jj] = tmp_err |
---|
805 | |
---|
806 | // solidAngle[ii][jj] = lat_err |
---|
807 | |
---|
808 | |
---|
809 | //solidAngle[ii][jj] = LargeAngleTransmissionCorr(trans,dtdist,xd,yd) //testing only |
---|
810 | endif |
---|
811 | |
---|
812 | jj+=1 |
---|
813 | while(jj<pixelsX) |
---|
814 | ii+=1 |
---|
815 | while(ii<pixelsX) |
---|
816 | |
---|
817 | //clean up waves |
---|
818 | |
---|
819 | Return(0) |
---|
820 | End |
---|
821 | |
---|
822 | |
---|
823 | |
---|
824 | |
---|
825 | // DIVIDE the intensity by this correction to get the right answer |
---|
826 | // TODO: |
---|
827 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
828 | // |
---|
829 | // |
---|
830 | Function V_LargeAngleTransmissionCorr(trans,dtdist,xd,yd,trans_err,err) |
---|
831 | Variable trans,dtdist,xd,yd,trans_err,&err |
---|
832 | |
---|
833 | DoAlert 0,"This has not yet been updated for VSANS" |
---|
834 | |
---|
835 | //angle dependent transmission correction |
---|
836 | Variable uval,arg,cos_th,correction,theta |
---|
837 | |
---|
838 | ////this section is the trans_correct() VAX routine |
---|
839 | // if(trans<0.1) |
---|
840 | // Print "***transmission is less than 0.1*** and is a significant correction" |
---|
841 | // endif |
---|
842 | // if(trans==0) |
---|
843 | // Print "***transmission is ZERO*** and has been reset to 1.0 for the averaging calculation" |
---|
844 | // trans = 1 |
---|
845 | // endif |
---|
846 | |
---|
847 | theta = atan( (sqrt(xd^2 + yd^2))/dtdist ) //theta at the input pixel |
---|
848 | |
---|
849 | //optical thickness |
---|
850 | uval = -ln(trans) //use natural logarithm |
---|
851 | cos_th = cos(theta) |
---|
852 | arg = (1-cos_th)/cos_th |
---|
853 | |
---|
854 | // a Taylor series around uval*arg=0 only needs about 4 terms for very good accuracy |
---|
855 | // correction= 1 - 0.5*uval*arg + (uval*arg)^2/6 - (uval*arg)^3/24 + (uval*arg)^4/120 |
---|
856 | // OR |
---|
857 | if((uval<0.01) || (cos_th>0.99)) |
---|
858 | //small arg, approx correction |
---|
859 | correction= 1-0.5*uval*arg |
---|
860 | else |
---|
861 | //large arg, exact correction |
---|
862 | correction = (1-exp(-uval*arg))/(uval*arg) |
---|
863 | endif |
---|
864 | |
---|
865 | Variable tmp |
---|
866 | |
---|
867 | if(trans == 1) |
---|
868 | err = 0 //no correction, no error |
---|
869 | else |
---|
870 | //sigT, calculated from the Taylor expansion |
---|
871 | tmp = (1/trans)*(arg/2-arg^2/3*uval+arg^3/8*uval^2-arg^4/30*uval^3) |
---|
872 | tmp *= tmp |
---|
873 | tmp *= trans_err^2 |
---|
874 | tmp = sqrt(tmp) //sigT |
---|
875 | |
---|
876 | err = tmp |
---|
877 | endif |
---|
878 | |
---|
879 | // Printf "trans error = %g\r",trans_err |
---|
880 | // Printf "correction = %g +/- %g\r", correction, err |
---|
881 | |
---|
882 | //end of transmission/pathlength correction |
---|
883 | |
---|
884 | return(correction) |
---|
885 | end |
---|
886 | |
---|
887 | |
---|
888 | // |
---|
889 | // TODO: |
---|
890 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
891 | // |
---|
892 | //test procedure, not called anymore |
---|
893 | Proc V_AbsoluteScaling(type,c0,c1,c2,c3,c4,c5) |
---|
894 | String type |
---|
895 | Variable c0=1,c1=0.1,c2=0.95,c3=0.1,c4=1,c5=32.0 |
---|
896 | Prompt type,"WORK data type",popup,"CAL;COR;SAM" |
---|
897 | Prompt c0, "Sample Transmission" |
---|
898 | Prompt c1, "Sample Thickness (cm)" |
---|
899 | Prompt c2, "Standard Transmission" |
---|
900 | Prompt c3, "Standard Thickness (cm)" |
---|
901 | Prompt c4, "I(0) from standard fit (normalized to 1E8 monitor cts)" |
---|
902 | Prompt c5, "Standard Cross-Section (cm-1)" |
---|
903 | |
---|
904 | Variable err |
---|
905 | //call the function to do the math |
---|
906 | //data from "type" will be scaled and deposited in ABS |
---|
907 | err = V_Absolute_Scale(type,c0,c1,c2,c3,c4,c5) |
---|
908 | |
---|
909 | if(err) |
---|
910 | Abort "Error in Absolute_Scale()" |
---|
911 | endif |
---|
912 | |
---|
913 | //contents are always dumped to ABS |
---|
914 | type = "ABS" |
---|
915 | |
---|
916 | String newTitle = "WORK_"+type |
---|
917 | DoWindow/F SANS_Data |
---|
918 | DoWindow/T SANS_Data, newTitle |
---|
919 | KillStrings/Z newTitle |
---|
920 | |
---|
921 | //need to update the display with "data" from the correct dataFolder |
---|
922 | //reset the current display type to "type" |
---|
923 | SVAR gCurDispType = root:Packages:NIST:VSANS:Globals:gCurDispType |
---|
924 | gCurDispType = Type |
---|
925 | |
---|
926 | V_fRawWindowHook() |
---|
927 | |
---|
928 | End |
---|
929 | |
---|
930 | // |
---|
931 | // TODO: |
---|
932 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
933 | // |
---|
934 | //s_ is the standard |
---|
935 | //w_ is the "work" file |
---|
936 | //both are work files and should already be normalized to 10^8 monitor counts |
---|
937 | Function V_Absolute_Scale(type,w_trans,w_thick,s_trans,s_thick,s_izero,s_cross,kappa_err) |
---|
938 | String type |
---|
939 | Variable w_trans,w_thick,s_trans,s_thick,s_izero,s_cross,kappa_err |
---|
940 | |
---|
941 | DoAlert 0,"This has not yet been updated for VSANS" |
---|
942 | |
---|
943 | //convert the "type" data to absolute scale using the given standard information |
---|
944 | //copying the "type" waves to ABS |
---|
945 | |
---|
946 | //check for existence of data, rescale to linear if needed |
---|
947 | String destPath |
---|
948 | //check for "type" |
---|
949 | if(WaveExists($("root:Packages:NIST:"+Type + ":data")) == 0) |
---|
950 | Print "There is no work file in "+type+"--Aborting" |
---|
951 | Return(1) //error condition |
---|
952 | Endif |
---|
953 | //check for log-scaling of the "type" data and adjust if necessary |
---|
954 | destPath = "root:Packages:NIST:"+Type |
---|
955 | NVAR gIsLogScale = $(destPath + ":gIsLogScale") |
---|
956 | if(gIsLogScale) |
---|
957 | Duplicate/O $(destPath + ":linear_data") $(destPath + ":data")//back to linear scale |
---|
958 | Variable/G $(destPath + ":gIsLogScale")=0 //the "type" data is not logscale anymore |
---|
959 | endif |
---|
960 | |
---|
961 | //copy "oldtype" information to ABS |
---|
962 | //overwriting out the old contents of the ABS folder (/O option in Duplicate) |
---|
963 | //copy over the waves data,vlegend,text,integers,reals(read) |
---|
964 | |
---|
965 | String oldType= "root:Packages:NIST:"+type //this is where the data to be absoluted is |
---|
966 | //copy from current dir (type) to ABS, defined by destPath |
---|
967 | Duplicate/O $(oldType + ":data"),$"root:Packages:NIST:ABS:data" |
---|
968 | Duplicate/O $(oldType + ":linear_data"),$"root:Packages:NIST:ABS:linear_data" |
---|
969 | Duplicate/O $(oldType + ":linear_data_error"),$"root:Packages:NIST:ABS:linear_data_error" |
---|
970 | // Duplicate/O $(oldType + ":vlegend"),$"root:Packages:NIST:ABS:vlegend" |
---|
971 | Duplicate/O $(oldType + ":textread"),$"root:Packages:NIST:ABS:textread" |
---|
972 | Duplicate/O $(oldType + ":integersread"),$"root:Packages:NIST:ABS:integersread" |
---|
973 | Duplicate/O $(oldType + ":realsread"),$"root:Packages:NIST:ABS:realsread" |
---|
974 | //need to save a copy of filelist string too (from the current type folder) |
---|
975 | SVAR oldFileList = $(oldType + ":fileList") |
---|
976 | //need to copy filelist string too |
---|
977 | String/G $"root:Packages:NIST:ABS:fileList" = oldFileList |
---|
978 | |
---|
979 | //now switch to ABS folder |
---|
980 | //make appropriate wave references |
---|
981 | WAVE data=$"root:Packages:NIST:ABS:linear_data" // these wave references point to the "type" data in ABS |
---|
982 | WAVE data_err=$"root:Packages:NIST:ABS:linear_data_error" // these wave references point to the "type" data in ABS |
---|
983 | WAVE data_copy=$"root:Packages:NIST:ABS:data" // just for display |
---|
984 | WAVE/T textread=$"root:Packages:NIST:ABS:textread" //that are to be directly operated on |
---|
985 | WAVE integersread=$"root:Packages:NIST:ABS:integersread" |
---|
986 | WAVE realsread=$"root:Packages:NIST:ABS:realsread" |
---|
987 | Variable/G $"root:Packages:NIST:ABS:gIsLogscale"=0 //make new flag in ABS folder, data is linear scale |
---|
988 | |
---|
989 | //do the actual absolute scaling here, modifying the data in ABS |
---|
990 | Variable defmon = 1e8,w_moncount,s1,s2,s3,s4 |
---|
991 | |
---|
992 | w_moncount = realsread[0] //monitor count in "type" |
---|
993 | if(w_moncount == 0) |
---|
994 | //zero monitor counts will give divide by zero --- |
---|
995 | DoAlert 0,"Total monitor count in data file is zero. No rescaling of data" |
---|
996 | Return(1) //report error |
---|
997 | Endif |
---|
998 | |
---|
999 | //calculate scale factor |
---|
1000 | Variable scale,trans_err |
---|
1001 | s1 = defmon/realsread[0] //[0] is monitor count (s1 should be 1) |
---|
1002 | s2 = s_thick/w_thick |
---|
1003 | s3 = s_trans/w_trans |
---|
1004 | s4 = s_cross/s_izero |
---|
1005 | |
---|
1006 | // kappa comes in as s_izero, so be sure to use 1/kappa_err |
---|
1007 | |
---|
1008 | data *= s1*s2*s3*s4 |
---|
1009 | |
---|
1010 | scale = s1*s2*s3*s4 |
---|
1011 | trans_err = realsRead[41] |
---|
1012 | |
---|
1013 | // print scale |
---|
1014 | // print data[0][0] |
---|
1015 | |
---|
1016 | data_err = sqrt(scale^2*data_err^2 + scale^2*data^2*(kappa_err^2/s_izero^2 +trans_err^2/w_trans^2)) |
---|
1017 | |
---|
1018 | // print data_err[0][0] |
---|
1019 | |
---|
1020 | // keep "data" in sync with linear_data |
---|
1021 | data_copy = data |
---|
1022 | |
---|
1023 | //********* 15APR02 |
---|
1024 | // DO NOt correct for atenuators here - the COR step already does this, putting all of the data one equal |
---|
1025 | // footing (zero atten) before doing the subtraction. |
---|
1026 | // |
---|
1027 | //Print "ABS data multiplied by ",s1*s2*s3*s4/attenFactor |
---|
1028 | |
---|
1029 | //update the ABS header information |
---|
1030 | textread[1] = date() + " " + time() //date + time stamp |
---|
1031 | |
---|
1032 | Return (0) //no error |
---|
1033 | End |
---|
1034 | |
---|
1035 | |
---|
1036 | // |
---|
1037 | // TODO: |
---|
1038 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
1039 | // |
---|
1040 | // |
---|
1041 | // match the attenuation of the RAW data to the "type" data |
---|
1042 | // so that they can be properly added |
---|
1043 | // |
---|
1044 | // are the attenuator numbers the same? if so exit |
---|
1045 | // |
---|
1046 | // if not, find the attenuator number for type |
---|
1047 | // - find both attenuation factors |
---|
1048 | // |
---|
1049 | // rescale the raw data to match the ratio of the two attenuation factors |
---|
1050 | // -- adjust the detector count (rw) |
---|
1051 | // -- the linear data |
---|
1052 | // |
---|
1053 | // |
---|
1054 | Function V_Adjust_RAW_Attenuation(type) |
---|
1055 | String type |
---|
1056 | |
---|
1057 | DoAlert 0,"This has not yet been updated for VSANS" |
---|
1058 | |
---|
1059 | WAVE rw=$("root:Packages:NIST:RAW:realsread") |
---|
1060 | WAVE linear_data=$("root:Packages:NIST:RAW:linear_data") |
---|
1061 | WAVE data=$("root:Packages:NIST:RAW:data") |
---|
1062 | WAVE data_err=$("root:Packages:NIST:RAW:linear_data_error") |
---|
1063 | WAVE/T tw = $("root:Packages:NIST:RAW:textRead") |
---|
1064 | |
---|
1065 | WAVE dest_reals=$("root:Packages:NIST:"+type+":realsread") |
---|
1066 | |
---|
1067 | Variable dest_atten,raw_atten,tol |
---|
1068 | Variable lambda,raw_atten_err,raw_AttenFactor,dest_attenFactor,dest_atten_err |
---|
1069 | String fileStr |
---|
1070 | |
---|
1071 | dest_atten = dest_reals[3] |
---|
1072 | raw_atten = rw[3] |
---|
1073 | |
---|
1074 | tol = 0.1 // within 0.1 atten units is OK |
---|
1075 | if(abs(dest_atten - raw_atten) < tol ) |
---|
1076 | return(0) |
---|
1077 | endif |
---|
1078 | |
---|
1079 | fileStr = tw[3] |
---|
1080 | lambda = rw[26] |
---|
1081 | // TODO access correct values |
---|
1082 | raw_AttenFactor = 1//AttenuationFactor(fileStr,lambda,raw_atten,raw_atten_err) |
---|
1083 | dest_AttenFactor = 1//AttenuationFactor(fileStr,lambda,dest_atten,dest_atten_err) |
---|
1084 | |
---|
1085 | rw[2] *= dest_AttenFactor/raw_AttenFactor |
---|
1086 | linear_data *= dest_AttenFactor/raw_AttenFactor |
---|
1087 | |
---|
1088 | // to keep "data" and linear_data in sync |
---|
1089 | data = linear_data |
---|
1090 | |
---|
1091 | return(0) |
---|
1092 | End |
---|
1093 | |
---|
1094 | // |
---|
1095 | // TODO: |
---|
1096 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
1097 | // |
---|
1098 | //************************ |
---|
1099 | //unused testing procedure, may not be up-to-date with other procedures |
---|
1100 | //check before re-implementing |
---|
1101 | // |
---|
1102 | Proc V_DIV_a_Workfile(type) |
---|
1103 | String type |
---|
1104 | Prompt type,"WORK data type",popup,"SAM;EMP;BGD;ADJ;" |
---|
1105 | |
---|
1106 | //macro will take whatever is in SELECTED folder and DIVide it by the current |
---|
1107 | //contents of the DIV folder - the function will check for existence |
---|
1108 | //before proceeding |
---|
1109 | |
---|
1110 | Abort "This has not yet been updated for VSANS" |
---|
1111 | |
---|
1112 | Variable err |
---|
1113 | err = V_DIVCorrection(type) //returns err = 1 if data doesn't exist in specified folders |
---|
1114 | |
---|
1115 | if(err) |
---|
1116 | Abort "error in V_DIVCorrection()" |
---|
1117 | endif |
---|
1118 | |
---|
1119 | //contents are NOT always dumped to CAL, but are in the new type folder |
---|
1120 | |
---|
1121 | String newTitle = "WORK_"+type |
---|
1122 | DoWindow/F VSANS_Data |
---|
1123 | DoWindow/T VSANS_Data, newTitle |
---|
1124 | KillStrings/Z newTitle |
---|
1125 | |
---|
1126 | //need to update the display with "data" from the correct dataFolder |
---|
1127 | //reset the current displaytype to "type" |
---|
1128 | String/G root:Packages:NIST:VSANS:Globals:gCurDispType=Type |
---|
1129 | |
---|
1130 | V_UpdateDisplayInformation(type) |
---|
1131 | |
---|
1132 | End |
---|
1133 | |
---|
1134 | |
---|
1135 | // |
---|
1136 | // TODO: |
---|
1137 | // -- DoAlert 0,"This has not yet been updated for VSANS" |
---|
1138 | // -- how is the error propagation handled? |
---|
1139 | // |
---|
1140 | //function will divide the contents of "workType" folder with the contents of |
---|
1141 | //the DIV folder + detStr |
---|
1142 | // all data is linear scale for the calculation |
---|
1143 | // |
---|
1144 | Function V_DIVCorrection(data,data_err,detStr,workType) |
---|
1145 | Wave data,data_err |
---|
1146 | String detStr,workType |
---|
1147 | |
---|
1148 | //check for existence of data in type and DIV |
---|
1149 | // if the desired data doesn't exist, let the user know, and abort |
---|
1150 | String destPath="" |
---|
1151 | |
---|
1152 | if(WaveExists(data) == 0) |
---|
1153 | Print "The data wave does not exist in V_DIVCorrection()" |
---|
1154 | Return(1) //error condition |
---|
1155 | Endif |
---|
1156 | |
---|
1157 | //check for DIV |
---|
1158 | // if the DIV workfile doesn't exist, let the user know,and abort |
---|
1159 | |
---|
1160 | WAVE/Z div_data = $("root:Packages:NIST:VSANS:DIV:entry:instrument:detector_"+detStr+":data") |
---|
1161 | if(WaveExists(div_data) == 0) |
---|
1162 | Print "The DIV wave does not exist in V_DIVCorrection()" |
---|
1163 | Return(1) //error condition |
---|
1164 | Endif |
---|
1165 | //files exist, proceed |
---|
1166 | |
---|
1167 | data /= div_data |
---|
1168 | |
---|
1169 | data_err /= div_data |
---|
1170 | |
---|
1171 | Return(0) |
---|
1172 | End |
---|
1173 | |
---|
1174 | |
---|
1175 | ////////////////////////// |
---|